This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||25 August 2006|
|PDF File Size:||7.48 Mb|
|ePub File Size:||1.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
Ensure the enclosure lid is secured by the 6 captivated screws. Page pkus P Tank shape Enter “1” for vertical cylinder with flat bottom. P Failsafe Timer Enter “. Enter a value slightly greater than the maximum vessel filling rate. This parameter cannot be reset by pressing values: There are 1 items available. Increase Measurement Response as required to stabilize the Reading, or install a stilling well contact Siemens Milltronics or your local distributor. Should a stable measurement still not be attainable, contact Siemens Milltronics or your local distributor.
Watch list is full. If the material presents a flat surface perpendicular to the transducer beam, select liquid. The Offset Correction amount is stored in P Select desired transducer point. Have one to sell? Scanning From this graph, determine airangrr Universal Volume calculation will provide the best results, and select the level vs volume breakpoints to be entered 32 max.
Milltronics airanger XPL plus Instruction Manual
Place the infrared programmer in the enclosure lid recess no wiring or fastening required. Page 36 Otherwise, when “fastback” is selected, the Pluss Level Advance is restricted, however the advance to the new material level when a valid measurement is made is immediate. Quick Start Parameters Enter the type of material to be monitored. Obstructed infrared interface, or Clean enclosure “docking bay” defective programmer.
Siemens SITRANS LU10 Ten-Point Ultrasonic Level Transmitter (Milltronics AiRanger XPL Plus)
Page 75 enters airanber change in value. Bic-ii Support Siemens Milltronics or your local xpp. To print the manual completely, please, download it. To select another transducer Press to set the point number to transducer. From this graph, determine which Universal Volume calculation will provide the best results, and select the level vs volume breakpoints to be entered 32 max. Numerous failsafe options for process control equipment activation.
Refer to associated product documentation.
To unLock, direct access this parameter and enter the value “”. When a Transducer P value is entered for Point 1 only, the Reading and other data is displayed airabger. Enter a negative value if Empty is below sea level. Please enter a valid postcode.
If Volume Conversion is not required, proceed to Reading Parameters. Select “short and long” to have short and long acoustic shots fired each time a vessel is scanned. For normal operation, return to the RUN mode.
Before using this feature, verify the: Not all components or their maximum quantity may be required. Distance Calculation Unless altered, the sound velocity used for the distance calculation is Repeat step 1 at least 5 times to overcome Echo Lock P and verify repeatability. With the Point Number field accessed, key in the Point Number to be reset 1 – No response to programmer.
Learn more about your rights as a buyer. A positive value is entered for a filling rate alarm, negative for emptying. To return an operator adjusted Parameter Value to the preset value, with the appropriate Point Number and Parameter Number displayed Page 52 To save a manual record: Learn More – opens in a new window or tab Any international postage and import charges are paid in part to Pitney Bowes Inc.